Device specified is known as Goodrich flexometer, but any other apparatus giving equivalent performance may be used. Predictions regarding the durability of rubbers in finished articles are possible by means of test results for the properties mentioned. Method not recommended for rubber having a hardness greater than 85 IRHD. - Part 1 of this International Standard: basic principles, part 2: rotary flexometer.
Status: WithdrawnPublication date: 1982-11
Edition: 1Number of pages: 7
Technical Committee: ISO/TC 45/SC 2 Testing and analysis
- ICS :
- 83.060 Rubber
ISO 4666-3:1982Stage: 95.99
Got a question?
Check out our FAQs
+41 22 749 08 88
Monday to Friday - 09:00-12:00, 14:00-17:00 (UTC+1)
Keep up to date with ISO
Sign up to our newsletter for the latest news, views and product information.